|Year : 2014 | Volume
| Issue : 3 | Page : 318-323
Evaluation of diagnostic accuracy of conventional and digital periapical radiography, panoramic radiography, and cone-beam computed tomography in the assessment of alveolar bone loss
Wilton Mitsunari Takeshita1, Lilian Cristina Vessoni Iwaki2, Mariliani Chicarelli Da Silva2, Renata Hernandes Tonin2
1 Department of Dentistry, Federal University of Sergipe, Aracaju, Brazil
2 Department of Dentistry State University of Maringá, Maringá, Brazil
|Date of Web Publication||2-Aug-2014|
Wilton Mitsunari Takeshita
Department of Dentistry, Federal University of Sergipe, Claudio Batista s/n, Cidade Universitária, Santo Antônio, Aracaju, Sergipe, CEP 49060 100
Source of Support: None, Conflict of Interest: None
| Abstract|| |
Background: To evaluate the diagnostic accuracy of different radiographic methods in the assessment of proximal alveolar bone loss (ABL). Materials and Methods: ABL, the distance between cement-enamel junction and alveolar bone crest, was measured in 70 mandibular human teeth - directly on the mandibles (control), using conventional periapical radiography with film holders (Rinn XCP and Han-Shin), digital periapical radiography with complementary metal-oxide semiconductor sensor, conventional panoramic, and cone-beam computed tomography (CBCT). Three programs were used to measure ABL on the images: Image tool 3.0 (University of Texas Health Sciences Center, San Antonio, Texas, USA), Kodak Imaging 6.1 (Kodak Dental Imaging 6.1, Carestream Health ® , Rochester, NY, USA), and i-CAT vision 1.6.20. Statistical analysis used ANOVA and Tukey's test at 5% significance level. Results: The tomographic images showed the highest means, whereas the lowest were found for periapical with Han-Shin. Controls differed from periapical with Han-Shin (P < 0.0001). CBCT differed from panoramic (P = 0.0130), periapical with Rinn XCP (P = 0.0066), periapical with Han-Shin (P < 0.0001), and digital periapical (P = 0.0027). Conventional periapicals with film holders differed from each other (P = 0.0007). Digital periapical differed from conventional periapical with Han-Shin (P = 0.0004). Conclusions: Conventional periapical with Han-Shin film holder was the only method that differed from the controls. CBCT had the closest means to the controls.
Keywords: Alveolar bone loss, cone-beam computerized tomography, conventional periapical radiography, diagnosis, panoramic radiography
|How to cite this article:|
Takeshita WM, Vessoni Iwaki LC, Da Silva MC, Tonin RH. Evaluation of diagnostic accuracy of conventional and digital periapical radiography, panoramic radiography, and cone-beam computed tomography in the assessment of alveolar bone loss. Contemp Clin Dent 2014;5:318-23
|How to cite this URL:|
Takeshita WM, Vessoni Iwaki LC, Da Silva MC, Tonin RH. Evaluation of diagnostic accuracy of conventional and digital periapical radiography, panoramic radiography, and cone-beam computed tomography in the assessment of alveolar bone loss. Contemp Clin Dent [serial online] 2014 [cited 2019 Apr 25];5:318-23. Available from: http://www.contempclindent.org/text.asp?2014/5/3/318/137930
| Introduction|| |
Alveolar bone height is the distance between the cement-enamel junction (CEJ) and the alveolar bone crest (ABC), along a line parallel to the long axis of the tooth. , This distance indicates whether there is alveolar bone loss (ABL) and bone alterations due to periodontal disease. Although studies showed considerable variation in this distance, from 0 to 3 mm, , a 2 mm distance is more often adopted as the norm for patients without periodontal disease. 
The diagnosis of periodontal diseases requires a careful clinical examination that takes into account the clinical history and associated signs and symptoms. In addition, radiographic examination is a fundamental tool to assess morphological and pathological changes in the periodontium, to assist the diagnosis, treatment planning, and prognosis of periodontal diseases. ,
The most used radiographic methods in the diagnosis of periodontal diseases are panoramic radiography, bitewings and periapical radiography, , despite their limitations - subjectivity of interpretation, image overlapping, and reduced sensitivity in detecting marginal bone changes. 
Constant technological development brought into dentistry other important imaging methods, such as digital radiography and cone-beam computed tomography (CBCT). Digital radiographic methods facilitate clinical practice, given that they eliminate chemical processing, reduce radiation exposure, and the images can be manipulated with the use of software.  This possibility of enhancing digital images optimizes diagnosis, unlike the static images of conventional films, which cannot be manipulated or enhanced. , However, as the geometry of digital images is still in two-dimension, the interpretation of intraoral and panoramic radiography do not benefit greatly from this aspect. 
Cone-beam computed tomography, on the other hand, provides images with excellent quality and accuracy, the structures can be seen in three-dimension at any angle, which allows analysis without distortion, and measurement of bone loss even of the buccal and lingual/palatal faces, a possibility not possible with other types of images. , However, du Bois et al.  assert that only about 3% of current studies in dentistry examine the applications of CBCT in periodontics. Does CBCT is so many good other image methods? CBCT may be unnecessary for interproximal ABL.
The aim of this in vitro study was to evaluate the diagnostic accuracy of conventional periapical radiography taken with film holders Rinn and Han-Shin, digital periapical radiography with complementary metal-oxide semiconductor sensor (CMOS), panoramic radiography, and CBCT in the measurement of ABL.
| Materials and Methods|| |
The study was approved by the Institutional Review Board/Independent Ethics Committee of the State University of Maringα (Protocol 0655142.7.0000.0104). The sample consisted of 70 teeth from 10 macerated human mandibles of the university's Department of Morphophysiology Sciences, each mandible with varied number of teeth.
The control method of measuring ABL consisted of determining the linear distance between the CEJ and the ABC on the interproximal surface of the teeth (the CEJ-ABC distance), with the use of a digital calliper accurate to 0.01 mm (Lee Tools ® , Rio de Janeiro, Brazil). The experimental methods consisted of measuring ABL using conventional periapical radiography with film holders (Rinn XCP and Han-Shin), digital periapical radiography with CMOS, panoramic radiography, and CBCT [Figure 1]. For better standardization were placed in small metal spheres in the CEJ and another peak of the ABC.
|Figure 1: (a) Cone-beam computed tomography image. (b) Digital periapical radiography with complementary metal-oxide semiconductor sensor. (c) Conventional periapical radiography with film holders Han-Shin. (d) Conventional periapical radiography with film holders Rinn XCP. (e) Panoramic radiography|
Click here to view
Conventional periapical radiographs were obtained using the paralleling technique with Rinn XCP film holder (Dentsply, USA) at a 40 cm focal length, and bisecting technique with Han-Shin film holder (Maquira ® , Maringα, Brazil) at a 20 cm focal length. Particular care was taken in order to minimize discrepancies among radiographs, by standardizing the geometrical projection of the X-rays, the alignment of the images, and contrast and density of the images. The bitewing technique was not performed because ABL was extensive in some teeth and could cut in the image.
Digital periapical radiographs were taken with the film holder provided by Kodak RVG 6100 digital radiography system (Carestream Health ® , Rochester, NY, USA). The digital periapical radiographs was measured using the equipment's own software (Kodak Dental Imaging 6.1, Carestream Health ® , Rochester, NY, USA). The files are saved according to the manufacturer's specifications.
Panoramic radiographs were obtained with a conventional X-ray machine (Orthoralix 9200 GENDEX, Dentsply ® , Des Plainnes, USA) using Kodak T-Mat G/RA films (15 × 30 cm, Carestream Health ® , New York, USA) with their metallic cassette and intensifying screen (Kodak Lanex Medium Extraoral Imaging Screens X-Omat). Using alignment lights, the mid sagittal plane of the mandibles was positioned perpendicularly to the ground and the occlusal plane parallel to the ground.
Calibration of panoramic radiography, conventional and digital periapical radiography was made using an aluminum step wedge (8 steps) with 1 mm increment and 5 mm thickness.
Conventional radiographs were processed in an automatic processor (Revell X-TEC, Londrina, Paranα, Brazil).
Cone-beam computed tomography images were obtained using an i-CAT scanner (Hatfield, PA, USA). For most favorable detailing of root structures, the study used a 14-bit grey scale, field of view of 6 cm, voxel of 0.125 mm, and 36.2 mAs exposure time. All radiographic and tomographic images were taken by the same operator, a specialist in dentomaxillofacial radiology. The files are saved according to the manufacturer's specifications.
Panoramic radiographs and conventional periapical radiographs were digitized using a scanner and a transparency adaptor (HP Scanjet G4050, Hewlett-Packard Company ® , Washington, USA) with optical scanning resolution of up to 4800 dpi. ,, Conventional radiographs were scanned at a resolution of 300 dpi and archived in uncompressed tagged-image file format (TIFF). The TIFF format is the best image format for digitized images, as if it were transformed into digital imaging and communications in medicine (DICOM) the details will be lost. ,
Alveolar bone loss from digitized conventional periapical radiographs and digitized panoramic radiographs was measured using Image Tool 3.00 software (University of Texas Health Sciences Center, San Antonio, Texas, USA), , according to Gόrdal et al.  the best image format using the Image tool software was the TIFF format. For CBCT images [Figure 2], ABL was measured using the equipment's own software (i-CAT Vision, Imaging Sciences International, LLC, Pennsylvania, USA). Measurements were made parallel to the long axis of the tooth, on the proximal surfaces. ,
File formats were determined by the image generated by the programs, except the conventional radiographs that had to be scanned format, that TIFF considered best for scanned images was used. , We chose to use the best file format and better resolution for each image type. All measurements were made by a single radiologist, experienced with the use of the mentioned programs, a specialist in dentomaxillofacial radiology, and were analyzed again with a 1 week interval between them. Intra-examiner reliability was verified by weighted kappa test (κW ). Intra-examiner coefficient was 0.843, the strength of agreement was very good according Altman  [Table 1], nevertheless the means of these two measurements were used for statistical analysis for greater reliability. Data were subjected to ANOVA and Tukey's test at 5% significance level, using GraphPad Prism 6.0 (GraphPad Software, San Diego, CA, USA) and Minitab 15.0 (Minitab Inc, State College PA, USA).
| Results|| |
The highest means of VBL were found for the actual measures using the calliper, the control method, and for the CBCT images, whereas the lowest ones values were for the images taken with the film holder Han-Shin [Table 2] and [Figure 1].
|Table 2: Mean and SD of ABL of different types of radiographic methods (n=70) |
Click here to view
Multiple comparisons of means using Tukey's test revealed that statistical differences [Table 3] were between control method and periapical using Han-Shin film holder (P < 0.001), but not between control and the other methods. The results of CBCT were different from those of panoramic radiographs (P = 0.0130), periapical with Rinn XCP film holder (P = 0.0066), periapical with Han-Shin film holder (P < 0.0001), and digital periapical radiographs (P = 0.0027). Periapical with film holders were different from each other (P = 0.0007). Digital periapical differed from periapical with Han-Shin film holder (P = 0.0004).
|Table 3: Tukey's test for multiple comparisons among different radiographic methods |
Click here to view
| Discussion|| |
Radiographic examination, along with clinical examination, is essential in the diagnosis of periodontal diseases, in determining their prognosis, and in assessing treatment outcome. ,,, Due to its relevance, this study aimed to evaluate the best radiographic method for measuring ABL.
In this study the results of periapical radiographs using the Rinn XCP film holders and Han-Shin were different from those Forsberg and Halse  and Hayashi et al. Although these authors concluded that both film holders show comparable accuracy if Han-Shin is correctly used and set at the right angle for the bisecting technique, they found that the paralleling technique is, in general, more accurate. In this study the results were similar to found by Rushton and Horner  and Coelho et al. 
Different from Persson et al.  and Langlois Cde et al.  who found that periapical radiographs and panoramic images showed similar accuracy, in the present study panoramic radiographs (mean 2.664) were more accurate to detect ABL than periapical using Han-Shin film holder (mean 2.224). However, compared with the control results the panoramic images were close, so indicating the panoramic as an initial test for the evaluation of ABL. In this respect, the authors , agree with results of this research.
In this manuscript, CBCT showed the most accurate ABL measurements, not statistically different from the control method. Similar results were found by Raichur et al.,  Vandenberghe et al.,  Grimard et al.,  and Mol and Balasundaram  who compared CBCT images with digital radiography in the detection of intrabony defects. The authors , used two measurements - from the CEJ to the bottom of the bone defect (CEJ-BD), and CEJ-ABC. They found that both CBCT and digital radiography had similar means for CEJ-ABC, whereas for CEJ-BD digital radiography showed a discrepancy of 1-2 mm from the actual means, with CBCT being more accurate to assess BD.
Mengel et al.  compared different radiographic methods for accuracy and quality of the representation of BDs. They found more accuracy for CBCT, the images in three plans with no overlapping or distortion. The same result found in the present manuscript.
Similarly, Georgescu et al.,  CBCT was significantly more accurate than panoramic images in this study. The scholars  evaluated CBCT and panoramic radiography, quantitatively and qualitatively, and concluded that CBCT images were more accurate to evaluate anterior mandibular area.
The comparison between CBCT with periapical radiographs showed, again, that CBCT is more accurate in detecting and locating ABL, as de Faria Vasconcelos et al.  also found. In contrast, Misch et al.,  who used CBCT, periapical radiography, periodontal probe, and an electronic calliper as the standard reference, found no difference between CBCT and periapical images in the detection of interproximal BDs. However, Misch et al.  also found that CBCT, but not periapical radiography, was able to detect buccal and lingual defects. As conventional radiography cannot assess the buccal and lingual surfaces, the present study opted to measure the interproximal surfaces, which allows the comparison among different radiographic methods.
Given the present research, it is clear that for interproximal BDs CBCT is not necessary given the fact that other techniques provide satisfactory accuracy except conventional periapical radiography using Han-Shin film holder.
| Conclusion|| |
Within the limitations of this study, compared with the control measurements, only conventional periapical radiography using Han-Shin film holder showed significant lower differences, whereas the values of CBCT were the closest to the control method.
| References|| |
|1.||Hausmann E. Radiographic and digital imaging in periodontal practice. J Periodontol 2000;71:497-503. |
|2.||du Bois AH, Kardachi B, Bartold PM. Is there a role for the use of volumetric cone beam computed tomography in periodontics? Aust Dent J 2012;57 Suppl 1:103-8. |
|3.||Mol A, Balasundaram A. In vitro cone beam computed tomography imaging of periodontal bone. Dentomaxillofac Radiol 2008;37:319-24. |
|4.||Wong BK, Leichter JW, Chandler NP, Cullinan MP, Holborow DW. Radiographic study of ethnic variation in alveolar bone height among New Zealand dental students. J Periodontol 2007;78:1070-4. |
|5.||Raichur PS, Setty SB, Thakur SL, Naikmasur VG. Comparison of radiovisiography and digital volume tomography to direct surgical measurements in the detection of infrabony defects. J Clin Exp Dent 2012;4:e43-7. |
|6.||Corbet EF, Ho DK, Lai SM. Radiographs in periodontal disease diagnosis and management. Aust Dent J 2009;54 Suppl 1:S27-43. |
|7.||Jeffcoat MK. Radiographic methods for the detection of progressive alveolar bone loss. J Periodontol 1992;63:367-72. |
|8.||Greenstein G. Subgingival irrigation - An adjunct to periodontal therapy. Current status and future directions. J Dent Hyg 1990;64:389-97. |
|9.||Mol A. Imaging methods in periodontology. Periodontol 2000 2004;34:34-48. |
|10.||Li G, Engström PE, Nasström K, Lü ZY, Sanderink G, Welander U. Marginal bone levels measured in film and digital radiographs corrected for attenuation and visual response: An in vivo study. Dentomaxillofac Radiol 2007;36:7-11. |
|11.||Mohan R, Singh A, Gundappa M. Three-dimensional imaging in periodontal diagnosis-Utilization of cone beam computed tomography. J Indian Soc Periodontol 2011;15:11-7. |
|12.||Tyndall DA, Rathore S. Cone-beam CT diagnostic applications: Caries, periodontal bone assessment, and endodontic applications. Dent Clin North Am 2008;52:825-41. |
|13.||Wolf B, von Bethlenfalvy E, Hassfeld S, Staehle HJ, Eickholz P. Reliability of assessing interproximal bone loss by digital radiography: Intrabony defects. J Clin Periodontol 2001;28:869-78. |
|14.||Persson RE, Tzannetou S, Feloutzis AG, Brägger U, Persson GR, Lang NP. Comparison between panoramic and intra-oral radiographs for the assessment of alveolar bone levels in a periodontal maintenance population. J Clin Periodontol 2003;30:833-9. |
|15.||Ordinola-Zapata R, Bramante CM, Duarte MH, Ramos Fernandes LM, Camargo EJ, de Moraes IG, et al. The influence of cone-beam computed tomography and periapical radiographic evaluation on the assessment of periapical bone destruction in dog's teeth. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2011;112:272-9. |
|16.||Bissoli CF, Takeshita WM, Castilho JC, Medici-filho E, Moraes ME. Digitalization of images in radiology: A new vision of future. Rev Odonto 2007;30:34-9. |
|17.||Gonçalves A, Nicoli GA, Zamperini CA, Hebling J, Gonçalves M. Effect of digitization parameters on periapical radiographic image quality with regard to anatomic landmarks. Rev Odontol UNESP 2011;40:25-9. |
|18.||Dove SB. UTHSCSA ImageTool: Version 3.0 Final is here. Available from: http://www.compdent.uthscsa.edu/dig/itdesc.html [Last accessed on 2014 Jan 27]. |
|19.||Takeshita WM, Vessoni Iwaki LC, Da Silva MC, Filho LI, Queiroz Ade F, Geron LB. Comparison of the diagnostic accuracy of direct digital radiography system, filtered images, and subtraction radiography. Contemp Clin Dent 2013;4:338-42. |
|20.||Gürdal P, Hildebolt CF, Akdeniz BG. The effects of different image file formats and image-analysis software programs on dental radiometric digital evaluations. Dentomaxillofac Radiol 2001;30:50-5. |
|21.||Lund H, Gröndahl K, Gröndahl HG. Cone beam computed tomography evaluations of marginal alveolar bone before and after orthodontic treatment combined with premolar extractions. Eur J Oral Sci 2012;120:201-11. |
|22.||Forsberg J, Halse A. Radiographic simulation of a periapical lesion comparing the paralleling and the bisecting-angle techniques. Int Endod J 1994;27:133-8. |
|23.||Altman DG. Practical Statistics for Medical Research. London: Chapman and Hall; 1991. |
|24.||Hayashi MM, Issa JP, Tiossi R, Pardini LC, Watanabe PC. Evaluation of the use of film holders by undergraduation students. Int J Odontostomatol 2007;1:109-14. |
|25.||Rushton VE, Horner K. A comparative study of radiographic quality with five periapical techniques in general dental practice. Dentomaxillofac Radiol 1994;23:37-45. |
|26.||Coelho RM, Sumas KS, Oliveira-Júnior IS, Manzi FR. Comparison of radiographic film holders considering periodontal aspects. Periodontia 2007;17:80-5. |
|27.||Langlois Cde O, Sampaio MC, Silva AE, Costa NP, Rockenbach MI. Accuracy of linear measurements before and after digitizing periapical and panoramic radiography images. Braz Dent J 2011;22:404-9. |
|28.||Vandenberghe B, Jacobs R, Yang J. Detection of periodontal bone loss using digital intraoral and cone beam computed tomography images: An in vitro assessment of bony and/or infrabony defects. Dentomaxillofac Radiol 2008;37:252-60. |
|29.||Grimard BA, Hoidal MJ, Mills MP, Mellonig JT, Nummikoski PV, Mealey BL. Comparison of clinical, periapical radiograph, and cone-beam volume tomography measurement techniques for assessing bone level changes following regenerative periodontal therapy. J Periodontol 2009;80:48-55. |
|30.||Mengel R, Kruse B, Flores-de-Jacoby L. Digital volume tomography in the diagnosis of peri-implant defects: An in vitro study on native pig mandibles. J Periodontol 2006;77:1234-41. |
|31.||Georgescu CE, Mihai A, Didilescu AC, Moraru R, Nimigean V, Nimigean VR, et al. Cone beam computed tomography as a method of quantitative and qualitative analysis of alveolar crest in the frontal mandibular area. Rom J Morphol Embryol 2010;51:713-7. |
|32.||de Faria Vasconcelos K, Evangelista KM, Rodrigues CD, Estrela C, de Sousa TO, Silva MA. Detection of periodontal bone loss using cone beam CT and intraoral radiography. Dentomaxillofac Radiol 2012;41:64-9. |
|33.||Misch KA, Yi ES, Sarment DP. Accuracy of cone beam computed tomography for periodontal defect measurements. J Periodontol 2006;77:1261-6. |
[Figure 1], [Figure 2]
[Table 1], [Table 2], [Table 3]
|This article has been cited by|
||Precision of cone beam CT to assess periodontal bone defects: a systematic review and meta-analysis
| ||Letícia Fernanda Haas,Glaucia Santos Zimmermann,Graziela De Luca Canto,Carlos Flores-Mir,Márcio Corrêa |
| ||Dentomaxillofacial Radiology. 2017; : 20170084 |
|[Pubmed] | [DOI]|
||Are dental and jaw changes more prevalent in a Brazilian population with sickle cell anemia?
| ||Halinna Larissa Cruz Correia de Carvalho,José Yagoh Saraiva Rolim,Érika Bárbara Abreu Fonseca Thomaz,Soraia de Fátima Carvalho Souza |
| ||Oral Surgery, Oral Medicine, Oral Pathology and Oral Radiology. 2017; |
|[Pubmed] | [DOI]|
||The emerging role of maxillofacial radiology in the diagnosis and management of patients with complex periodontitis
| ||William C. Scarfe,Bruno Azevedo,Lucas R. Pinheiro,Menik Priaminiarti,Marcelo A. O. Sales |
| ||Periodontology 2000. 2017; 74(1): 116 |
|[Pubmed] | [DOI]|
||3D CBCT Assessment of Soft Tissue Calcification
| ||Santosh R Patil,Mohammad Khursheed Alam,Keita Moriyama,Saeka Matsuda,Masahito Shoumura,Naoto Osuga |
| ||Journal of Hard Tissue Biology. 2017; 26(3): 297 |
|[Pubmed] | [DOI]|
||Comparison of periodontal evaluation by cone-beam computed tomography, and clinical and intraoral radiographic examinations
| ||Wenjian Zhang,Shazia Rajani,Bing-Yan Wang |
| ||Oral Radiology. 2017; |
|[Pubmed] | [DOI]|
||A six-site method for the evaluation of periodontal bone loss in cone-beam CT images
| ||Yu-Jiao Guo,Zhi-pu Ge,Ruo-han Ma,Jian-xia Hou,Gang Li |
| ||Dentomaxillofacial Radiology. 2016; 45(1): 20150265 |
|[Pubmed] | [DOI]|